EasyPulse Portable Oxygen Concentrator 3 Liter
Mfr Model# PM4130P
Condition: Brand new with manufacturer's warranty
5 Year Manufacturer's Warranty
FREE Nasal Cannulas and Backpack included
The EasyPulse Portable Oxygen Concentrator (POC) is the perfect product to help you reduce costly cylinder deliveries and allow for your patients ease of mobility at the same time.
Most used oxygen conserver on the market with an oxygen concentrator to produce the EasyPulse POC.
The EasyPulse Portable Oxygen Concentrator has an easy user interface.
Lightest 3 Liter portable concentrator on the market, weighing only 6.6 pounds in a package that is only 10.1 in high x 6.5 in wide x 4.5 in deep.
Low Maintenance: Only one filter that is easily cleaned by the user.
Excellent Battery Life: Standard internal battery lasts over 3 hours at 2LPM.
The unit can also be used with the AC cord or with the DC car adapter.
Height: 10.10 in (25.7 cm); Width: 6.5 in (16.5 cm); Depth: 4.5 in (11.4 cm)
AC: 100 - 240 VAC (1.0 A @ 120 VAC) DC: 11 - 16 VDC (5.0 A @ 12 VDC)
Internal Battery Average Charge Time: 4 Hours
Oxygen Concentration: 87 to 95% @ 3 l/min setting within 4 to 6 minutes
Environmental Conditions: Temperature Range: 41*F to 104*F (5*C to 40*C); Storage Temperature: - 4*F to 140*F (- 20*C to 60*C); Relative Humidity: Up to 95% Non-condensing
Altitude: Sea Level to 14,000 ft. (4,267 m)
Outlet Temperature: Within 2*F (1.1*C) of ambient
Noise level: 40 dBA @ setting 2
The Precision Medical EasyPulse POC 3 is a lightweight, easy to use portable oxygen concentrator that offers both reliability and quality. Weighing approximately 5 pounds with the internal battery, the EasyPulse delivers oxygen therapy for up to 5.5 hours (11 hours with the external battery)—making it the perfect companion for any trip. With a simple control panel, users can easily check battery level, change flow settings, and monitor alarms.
EasyPulse POC 3 is a premium product that can provide therapeutic oxygen consistently over a very long period of time. An easy-to-use reliable option for patients who demand quality and reliability. Just press the button and enter the prescribed settings on the control panel to start the operation. Enjoy the freedom to sleep comfortably at home or when you travel by air, water, or land. This highly portable oxygen concentrator provides an unlimited supply of therapeutic oxygen to patients.
The EasyPulse POC 3 from Precision Medical delivers a pulse dose of the oxygen it concentrates. As the user inhales, the machine sends a pulse (also called a “bolus”) of oxygen through the tubing to the nose. This portable oxygen concentrator provides three pulse dose settings ranging from 240 to 520 mL/min, ensuring that users will get the oxygen therapy they need wherever they are. Each set is indicated by a numbered button on the control panel. An LED light under each number will flash as oxygen is delivered to the user. These settings are based on your physician’s prescription and should not be changed prior to consulting your doctor.
To get the most out of your oxygen therapy, it is recommended that you do not use oxygen tubing greater than 7 feet in length. Longer tubing may prevent the breath sensors in the device from generating the bolus required for your therapy.
The EasyPulse POC3 oxygen concentrator has an internal battery that provides up to 5.5 hours of life. During checkout, customers can add the external battery pack for an extra 5.5 hours of battery for a total of 11 hours (at a pulse dose setting of 1). The external battery cannot be charged while in use with the concentrator. To view the battery life of the internal and external batteries based on flow settings, refer to the chart below.
The battery indicator on the control panel of the device displays the remaining battery life of the unit. When the battery reaches a 15% charge, the LED indicator will turn red and the concentrator will beep once every minute. With 5% battery life remaining, the red LED will begin to flash and the unit will beep once every 20 seconds until it is connected to a power source.
The EasyPulse POC3 comes with both AC and DC power cords. Both will charge the internal battery while the device is plugged in.
EasyPulse POC 3 Features
EasyPulse is a renowned brand in the world of portable oxygen concentrators. Patients highly recommend EasyPulse products, and the PM4130 is no exception. It uses high-quality molecular sieve beds which are covered under a 3-year warranty. Whether you want to buy this for yourself or for treating patients, you can use this device. With the innovative Vacuum Pressure Swing Adsorption (VPSA) technology, each sieve bed gets rejuvenated every cycle. As a result, you get clean and pure oxygen with every breath.
Enjoy long battery duration with Controlled Minute Volume (CMV) technology that provides therapeutic oxygen in the first third of the inspiratory phase for maximum efficiency, regardless of breath rate.
Ease of Access
The device is easy to use. Just turn it on and turn the dial to the desired flow to begin operation. There is only one button and a dial for operation.
Its compact design ensures that you can take it with you wherever you want. One of the smallest POCs in the market, it weighs only 4.9 lbs.
Long Lasting Battery
The EasyPulse POC 3 provides power for one sleeping session with its small-sized yet powerful battery. The combination of the internal and external battery can make it last for 11 hours as each battery powers the device for a duration of 5.5 hours. The external battery is an additional accessory that has to be bought separately.
Each unit is tested rigorously to ensure its durability and also the level of oxygen purity that it can deliver. Each device is guaranteed to deliver 95.5% of oxygen purity which is considered to be the highest level for therapy.
Safety & Troubleshooting
The Precision Medical EasyPulse POC3 comes with built-in alarms to alert users when something is not working properly. An alarm indicator LED is located on the bottom-right of the control panel. A green LED indicates that the oxygen concentrator is functioning properly and ready to use. A yellow LED indicates a medium-priority alert while a red LED indicates a high priority alert.
Examples of alarms include: no breath detected, breath rate exceeded, abnormal functionality, or device failure. To troubleshoot an alarm, reference the user manual. If a solution cannot be determined, contact our customer care team for assistance.
Cleaning & Maintenance
The POC3 requires very little maintenance. As always, do not attempt to open or repair your portable oxygen concentrator on your own. If the device is not functioning properly, consult our customer care team for assistance.
When cleaning the EasyPulse portable oxygen concentrator, make sure it is disconnected from any external power source. Simply wipe it down with a damp cloth and a mild detergent. Let the unit air dry completely before plugging back in and using it.
To replace the inlet filter, pull the filter tab from the back of the device. Wash the filter thoroughly with water and a mild detergent. Do not reinsert the filter while it is still wet.
It is FAA-approved for use during commercial flights. We recommend contacting your airline prior to your trip to ensure that all arrangements for you and your portable oxygen concentrator are made in advance.
- Go through the user manual before using this device.
- This device should only be used by either an adult or by a child under the supervision of an adult.
- If the patient can't see or hear alarms or communicate discomfort, someone should monitor the use of the POC
- Never rotate the knob below the "Off" position or over "Setting 3" otherwise it will damage the knob.
- Use this device only at the settings prescribed by your doctor.
- Check the nasal cannula for any kinking or obstruction before each use.
- This POC doesn't support the use of humidifier bottles.
- The use of low flow, pediatric oxygen masks, and nasal cannulas is prohibited.
- Check for any visible damage to the device before using it.
- Never use this device without the inlet filter or with a wet inlet filter.
A carry bag is provided on the purchase of the Precision Medical EasyPulse portable oxygen concentrator. It provides ease in transporting, prevention of movement from vibration, and protection to the device from minor mishaps. This carry bag doesn't prevent the device from damage from accidents or abuse or major falls nor is it waterproof.
To use the carry bag for traveling with this device, make sure of the following:
- The bag is clean and empty.
- The mesh opening along the bottom of the bag is aligning with the bottom exhaust vents.
- Aligning the mesh with the exhaust vents makes sure that the POC remains cool and operates properly.
- Keep the exhaust vents facing away from the body.
- Ensure the outside of the bag does not get hot. If this happens then remove the POC and check to make sure no vent is being obstructed.
- Never cover vents as they provide the required ventilation for the device.
Internal Battery Pack
Precision Medical EasyPulse portable oxygen concentrator works with a lithium-ion internal rechargeable battery pack. You can check the charge level of the internal battery by looking at the Battery Level Indicator in the upper right corner of the control panel. It continuously shows the charge level. LEDs indicate the battery level. For instance, a solid red LED will illuminate with a brief audible beep sound once a minute when the charge level drops below 15%. When the charge level of the battery drops below 5%, the red LED will flash and a beep will be heard every 20 seconds indicating that the battery is about to deplete and needs to be recharged.
Proper Use: The battery life can be prolonged by frequently charging the battery instead of waiting for the battery to be fully depleted.
Storage: You can store the internal lithium-ion battery if you want to use other power sources to operate the POC. Simply charge the battery to around 50% before storing it.